Describe the principles of measurement of arterial haemoglobin O2 saturation using a pulse oximeter. (60% of marks) Outline the limitations of this technique. (40% of marks)
A lack of understanding of the physics behind pulse oximetry was a common area of weakness amongst most candidates. Candidates were expected to mention the underlying principle of the “Beer Lambert” Law, absorption spectra and that the differential absorption of light at different wavelengths by different haemoglobin species is used to determine the fractions of haemoglobin types. Limitations should include mention of errors due to calibration as well as sources of false positive and false negative readings.
Sinex, James E. "Pulse oximetry: principles and limitations." The American journal of emergency medicine 17.1 (1999): 59-66.
Jubran, Amal. "Pulse oximetry." Critical care 3.2 (1999): R11.
Jubran, Amal. "Pulse oximetry." Applied Physiology in Intensive Care Medicine 1. Springer, Berlin, Heidelberg, 2012. 51-54.
Jubran, Amal. "Pulse oximetry." Critical Care 19.1 (2015): 272.
Kamat, Vijaylakshmi. "Pulse oximetry." Indian J Anaesth 46.4 (2002): 261-8.
Kyriacou, Panayiotis, Karthik Budidha, and Tomas Y. Abay. "Optical techniques for blood and tissue oxygenation." Encyclopedia of Biomedical Engineering, ed R. Narayan (Oxford: Elsevier) (2019): 461-472.
Severinghaus, John W., and Shin O. Koh. "Effect of anemia on pulse oximeter accuracy at low saturation." Journal of clinical monitoring 6.2 (1990): 85-88.
Fluck, Robert R., et al. "Does ambient light affect the accuracy of pulse oximetry?." Respiratory care 48.7 (2003): 677-680.
Amar, David, et al. "Fluorescent light interferes with pulse oximetry." Journal of clinical monitoring 5.2 (1989): 135-136.
Block, Frank E. "Interference in a pulse oximeter from a fiberoptic light source." Journal of clinical monitoring 3.3 (1987): 210-211.
American Association for Respiratory Care. AARC Clinical Practice Guideline: Pulse oximetry. Respir Care 1991;36(12):1406–1409
Ralston, A. C., R. K. Webb, and W. B. Runciman. "Potential errors in pulse oximetry: I. Pulse oximeter evaluation." Anaesthesia 46.3 (1991): 202-206.
Webb, R. K., A. C. Ralston, and W. B. Runciman. "Potential errors in pulse oximetry: II. Effects of changes in saturation and signal quality." Anaesthesia 46.3 (1991): 207-212.
Singh, Anupam Kumar, et al. "Comparative evaluation of accuracy of pulse oximeters and factors affecting their performance in a tertiary intensive care unit." Journal of clinical and diagnostic research: JCDR 11.6 (2017): OC05.
Severinghaus, J. W. "Pulse oximetry uses and limitations." ASA Convention. 1989.
Ralston, A. C., R. K. Webb, and W. B. Runciman. "Potential errors in pulse oximetry III: effects of interference, dyes, dyshaemoglobins and other pigments." Anaesthesia 46.4 (1991): 291-295.
Severinghaus, John W. "Takuo Aoyagi: discovery of pulse oximetry." Anesthesia & Analgesia 105.6 (2007): S1-S4.
Tremper, Kevin K. "Pulse oximetry." Chest 95.4 (1989): 713-715.
Feiner, John R., John W. Severinghaus, and Philip E. Bickler. "Dark skin decreases the accuracy of pulse oximeters at low oxygen saturation: the effects of oximeter probe type and gender." Anesthesia & Analgesia 105.6 (2007): S18-S23.
Pologe, Jonas A. "Pulse oximetry: technical aspects of machine design." International anesthesiology clinics 25.3 (1987): 137-153.
Webster, John G. Design of pulse oximeters. CRC Press, 1997.