Describe the essential components of an ECG monitor (80% of marks). Outline the methods employed to reduce artefact (20% of marks).
The ECG device detects and amplifies the small electrical changes on the skin that are caused
when the heart muscle depolarizes (0.5 – 2 mV). This is reflected as rises and falls in the
voltage between two electrodes placed either side of the heart which is displayed either on a
screen or on paper. Usually more than 2 electrodes are used and they can be combined into a
number of pairs (For example: Left arm (LA), right arm (RA) and left leg (LL) electrodes form
the three pairs LA+RA, LA+LL, and RA+LL). The output from each pair is known as a lead.
Each lead is said to look at the heart from a different angle.
Electrodes are commonly made of silver or silver chloride components that are attached to the
main unit of the machine. Most ECG machines use 12 electrodes. Better answers made
mention of the two lead types: unipolar and bipolar.
Methods to reduce artefact include improving signal detection (conductive paste, skin
preparation (dry, no hair, etc.)) and minimizing external electrostatic forces (common earthed
environment, diathermy, etc.,) or patient environment (avoid shivering). The amplifier has three essential functions: High input impedance so as to minimize signal loss
and reject interference (50 – 60 Hz), differential amplification, (to amplify the potential
difference detected by the skin electrodes), and high common mode rejection (e.g. > 50Hz) to
aid eliminating muscle artefact or electrical interference from the power grid.).
Vector analysis is not a component of the ECG machine and so was not required to answer the question.
de Luna, Antoni Bayés. Chapter 2; "The ECG Curve: What Is It and How Does It Originate?" in: ECGs for beginners. John Wiley & Sons, 2014.
Dupre, Anthony, Sarah Vincent, and Paul A. Iaizzo. "Basic ECG theory, recordings, and interpretation." Handbook of cardiac anatomy, physiology, and devices. Humana Press, 2005. 191-201.
Reisner, Andrew T., Gari D. Clifford, and Roger G. Mark. "The physiological basis of the electrocardiogram." Advanced methods and tools for ECG data analysis 1 (2006): 25.
Hurst, J. Willis. "Naming of the waves in the ECG, with a brief account of their genesis." Circulation 98.18 (1998): 1937-1942.
Geselowitz, David B. "Dipole theory in electrocardiography." The American journal of cardiology 14.3 (1964): 301-306.
Crouch, Catherine Hirshfeld, and John W. Hirshfeld Jr. "Teaching the electrical origins of the electrocardiogram: An introductory physics laboratory for life science students." American Journal of Physics 88.7 (2020): 526.
Bailey, James J., et al. "Recommendations for standardization and specifications in automated electrocardiography: bandwidth and digital signal processing. A report for health professionals by an ad hoc writing group of the Committee on Electrocardiography and Cardiac Electrophysiology of the Council on Clinical Cardiology, American Heart Association." Circulation 81.2 (1990): 730-739.
Wilson, Frank N. "The distribution of the potential differences produced by the heart beat within the body and at its surface." American Heart Journal 5.5 (1930): 599-616.
HOLT JR, JOHN H., et al. "A study of the human heart as a multiple dipole electrical source: I. Normal adult male subjects." Circulation 40.5 (1969): 687-696.
Hobbie, Russell K., and Bradley J. Roth. "The Exterior Potential and the Electrocardiogram." Intermediate Physics for Medicine and Biology. Springer, Cham, 2015. 185-212.
Hobbie, Russell K. "The electrocardiogram as an example of electrostatics." American Journal of Physics 41.6 (1973): 824-831.
Malmivuo, Jaakko, and Robert Plonsey. Bioelectromagnetism: principles and applications of bioelectric and biomagnetic fields. Oxford University Press, USA, 1995.
Vieau, Sarah, and Paul A. Iaizzo. "Basic ECG theory, 12-lead recordings, and their interpretation." Handbook of Cardiac Anatomy, Physiology, and Devices. Springer, Cham, 2015. 321-334.https://bradscholars.brad.ac.uk/bitstream/handle/10454/16940/Youseffi_JTECE.pdf?sequence=1&isAllowed=y
Abdul Jamil, M. M., et al. "Electrocardiograph (ECG) circuit design and software-based processing using LabVIEW." (2017). Journal of Telecommunication, Electronic and Computer Engineering.Vol. 9 No. 3-8
Wilson, Frank N., A. Garrard Macleod, and Paul S. Barker. "The accuracy of Einthoven's equation." American Heart Journal 7.2 (1931): 203-206.
Wilson, Frank N., et al. "Electrocardiograms that represent the potential variations of a single electrode." American Heart Journal 9.4 (1934): 447-458.
Kligfield, P., et al. "American Heart Association Electrocardiography and Arrhythmias Committee, Council on Clinical Cardiology; American College of Cardiology Foundation." Heart Rhythm Society. Recommendations for the standardization and interpretation of the electrocardiogram: part I: the electrocardiogram and its technology a scientific statement from the American Heart Association Electrocardiography and Arrhythmias Committee, Council on Clinical Cardiology (2007): 1109-1127.
Bosznai, István, Ferenc Ender, and Hunor Sántha. "Web server based remote health monitoring system." 2009 32nd International Spring Seminar on Electronics Technology. IEEE, 2009.
Grassini, S. "Electrochemical Impedance Spectroscopy (EIS) for the in-situ analysis of metallic heritage artefacts." Corrosion and Conservation of Cultural Heritage Metallic Artefacts. Woodhead Publishing, 2013. 347-367.
Spach, Madison S., et al. "Skin-electrode impedance and its effect on recording cardiac potentials." Circulation 34.4 (1966): 649-656.
Davies, Alan. "Recognizing and reducing interference on 12-lead electrocardiograms." British journal of nursing 16.13 (2007): 800-804.
Patel, Santosh I., et al. "Equipment-related electrocardiographic artifacts: causes, characteristics, consequences, and correction." The Journal of the American Society of Anesthesiologists 108.1 (2008): 138-148.
Selvan, R. Barani, et al. "Earthing defect: A cause for unstable hemodynamics." Annals of cardiac anaesthesia 15.1 (2012): 47.
Somerville, T. "Isolation amps hike accuracy and reliability." Electronic Design 38.4 (1990): 127-132.