This is a simple process. You pump anticoagulated blood through a hemoperfusion cartridge. Empirically, it seems a good blood flow rate is about 300ml/min. A rising filter pressure indicates that a clot is forming in the filter.
Haemoperfusion is defined as "the extracorporeal procedure in which the anticoagulated patient’s blood passes through a volume of adsorbent material"
Adsorption is the deposition of molecules on the surface of a medium, rather than within it (which would be aBsorption).
The adsoprtion medium needs to have several favourable properies:
Charcoal = 300g of activated charcoal
Resin = 650 g of XAD polystyrene resin
Primed and flushed with normal saline.
Each cartridge lasts about 4 hours.
Obviously, it is not reusable.
Relevant features to make note of :
Filth getting sucked back into the patient:
If something is so highly protein bound that it is not easily dialysed, then one can assume it will be distributed into the tissue proteins as well as circulating proteins.
The result is that the toxin will redistribute back into the "clean" blood which goes back into the patient, and thus multiple sessions of hemoperfusion may be required to wash out all of the tissue-bound toxin.
For a definitive treatment of all of this, you ought to pay homage to the gigantic and all-encompassing "Critical Care Nephrology" by Ronco Bellomo and Kellum (2009).
The Gambro and Fresenius websites have also been an excellent source of information.
Rafael Ponikvar, "Hemoperfusion" in: Critical Care Nephrology (2009) p.1535
Nikolas Harbord, Steven J. Gruber, Donald A. Feinfeld, and James Frank Winchester "Hemodialysis, Hemofiltration, and Hemoperfusion in Acute Intoxication and Poisoning" in: Critical Care Nephrology (2009) p.919
Gil, H-W., et al. "Clinical outcome of hemoperfusion in poisoned patients."Blood purification 30.2 (2010): 84-88.
Winchester, James F. "Complications of Hemoperfusion." In: Complications of Dialysis (2000): p.127.
Fennimore, J., J. C. Kolthammen, and S. M. Lang. "Evaluation of hemoperfusion systems: in-vitro methods related to performance and safety."Artificial Organs (1977). - this article is not available anywhere, even as an abstract!
Cruz, Dinna N., et al. "Early use of polymyxin B hemoperfusion in abdominal septic shock: the EUPHAS randomized controlled trial." Jama 301.23 (2009): 2445-2452.